|Summary:||ASTERISK-02080: E1 stopped working in recent CVS.|
|Date Opened:||2004-07-22 04:05:19||Date Closed:||2004-09-25 02:37:47|
|Description:||Since a recent update of Zaptel from CVS, the E1 connected to that box through a TE410P has stopped working. The D-channel simply no longer comes up. "pri intense debug span 1" shows one of these about every second when it goes wrong:|
> Unnumbered frame:
> SAPI: 00 C/R: 0 EA: 0bug span
> TEI: 000 EA: 1
> M3: 3 P/F: 1 M2: 3 11: 3 [ SABME (set asynchronous balanced mode extended) ]
> 0 bytes of data
Reverting to the Zaptel-version dated 07/14/04 from CVS fixes this problem.
****** ADDITIONAL INFORMATION ******
Dual Xeon 2.4GHz.
Relevant parts of config-files:
channel => 1-15,17-31
|Comments:||By: adomjan (adomjan) 2004-07-22 18:36:58|
I tried upgrade my system from e100p to te405p, and failed with the same problem.
cvs: 07/21/04, so I changed back to e100p.
By: Mark Spencer (markster) 2004-07-22 19:45:54
Can you identify specifically which version introduced the problem?
By: Mark Spencer (markster) 2004-07-22 19:48:47
Nevermind, found it. It's fixed in CVS, if not, just reopen.